Scythe Kama Thermo
Published: 1st April 2007 | Source: Scythe | Price: |
The main aim of of this part of the review is purely to assess the Kama Thermo's accuracy according to a given tolerance...The tolerance in this case, happens to be +/- 1 degree C, according to Scythe's specifications. Because the Kama Thermo is essentially a digital thermometer, I shall be assessing it's accuracy using a regular glass-mercury type thermometer during the performance tests.
The tests I have chosen to assess the performance of the Kama Thermo are those of the ambient air temperature in my office and the temperatures present from within my PC case. Both the glass-mercury thermometer and Kama Thermo temperature probe will be positioned in exactly the same area as each other to prevent any reading ambiguities. The mercury-glass thermometer was allowed 5 mins to aclimatise to the surrounding temperatures.
Now that we've looked at the Kama Thermo in detail and assessed it's performance, let head onto the conclusion page to see just how well it scored in today's review.